ASTM-B588 › Standard Test Method for Measurement of Thickness of Transparent or Opaque Coatings by Double-Beam Interference Microscope Technique (Withdrawn 2016)
Show Complete Document History
The following bibliographic material is provided to assist you with your purchasing decision:
1.1 This test method covers the measurement of the thickness of transparent metal oxide and metallic coatings by utilizing a double-beam interference microscope.
1.2 The test method requires that the specimen surface or surfaces be sufficiently mirrorlike to form recognizable fringes.
1.3 This test method can be used nondestructively to measure 1 to 10μ m thick transparent coatings, such as anodic coatings on aluminum. The test method is used destructively for 0.1 to 10 μm thick opaque coatings by stripping a portion of the coating and measuring the step height between the coating and the exposed substrate. The stripping method can also be used to measure 0.2 to 10 μm thick anodic coatings on aluminum.
1.4 The test method is usable as a reference method for the measurement of the thickness of the anodic film on aluminum or of metallic coatings when the technique includes complete stripping of a portion of the coating without attack of the substrate. For anodic films on aluminum, the thickness must be greater than 0.4 μm; the uncertainty can be as great as 0.2 μm. For metallic coatings, the thickness must be greater than 0.25 μm; the uncertainty can be as great as 0.1 μm.
1.5 The values stated in SI units are to be regarded as standard. No other units of measurement are included in this standard.
1.6 This standard does not purport to address all of the safety concerns, if any, associated with its use. It is the responsibility of the user of this standard to establish appropriate safety and health practices and determine the applicability of regulatory limitations prior to use.
Significance and Use
The thickness of a coating is often critical to its performance.
For some coating-substrate combinations, the interference microscope method is a reliable method for measuring coating thickness.
This test method is suitable for specification acceptance.
Aluminum coatings; Anodic coatings; Anodic coatings--Al/Al alloys); Chromium electroplating; Coating thickness; Destructive testing--coatings; Double-beam interference microscopy; Electrodeposited coatings; Metallic coatings; Microscopic examination--metallic/inorganic coatings; Opacity/opaque materials; Steel--electrodeposited coatings; Thickness--electrodeposited coatings; Transparent coatings; ICS Number Code 17.040.20 (Properties of surfaces)
To find similar documents by ASTM Volume:
To find similar documents by classification:
This document comes with our free Notification Service, good for the life of the document.
This document is available in either Paper or PDF format.
Customers who bought this document also bought:
Anodic Coatings for Aluminum and Aluminum Alloys
Standard Test Method for Measurement of Thickness of Anodic Coatings on Aluminum and of Other Nonconductive Coatings on Nonmagnetic Basis Metals with Eddy-Current Instruments
Anodizing of aluminium and its alloys - Determination of breakdown voltage and withstand voltage
1988 R10 EDITION
Nov. 1, 2010