ASTM-F2024 › Standard Practice for X-ray Diffraction Determination of Phase Content of Plasma-Sprayed Hydroxyapatite Coatings
Show Complete Document History
The following bibliographic material is provided to assist you with your purchasing decision:
1.1 This practice is for the determination, by the Reference Intensity Ratio External Standard Method, of the percent by weight of the crystalline phases, hydroxyapatite (HA), beta-(whitlockite) tricalcium phosphate (β-TCP), and calcium oxide (CaO) in coatings deposited upon metallic substrates by plasma-spraying hydroxyapatite.
1.2 A major component in plasma-sprayed HA coatings other than HA is expected to be amorphous calcium phosphate (ACP). Crystalline components other than HA that may be present include alpha- and beta- (whitlockite) tricalcium phosphates, tetracalcium phosphate (TTCP), calcium oxide, and calcium pyrophosphates. Quantification of the minor crystalline components has proven to be very unreliable due to extreme overlap and confounding of X-ray diffraction peaks. Therefore, this practice addresses the quantification of only HA, β-TCP, and CaO.
1.3 This practice was developed for plasma-sprayed HA coatings with HA contents of at least 50 % of the total coating. It is recognized that the analysis of the crystalline components uses diffraction from regions of the pattern that also includes a small contribution from the amorphous component. However, within the limits of applicability of this practice, the effect of such interference is believed to be negligible.
1.4 The coating analyzed shall be produced and processed under equivalent manufacturing conditions to that on the device of interest.
1.5 This practice requires the use of monochromated copper Kα radiation and flat samples.
1.6 The values stated in SI units are to be regarded as standard. No other units of measurement are included in this standard.
1.7 This standard does not purport to address all of the safety concerns, if any, associated with its use. It is the responsibility of the user of this standard to establish appropriate safety and health practices and determine the applicability of regulatory limitations prior to use.
Significance and Use
3.1 Calcium phosphate coatings have been shown in animal and clinical studies to be biocompatible and to enhance the early attachment of bone to implant surfaces (see Refs. )
3.2 It is believed that the form of calcium phosphate ceramic and its purity with respect to secondary crystalline phases and amorphous material have an effect on its physical, mechanical, and biological properties. However, no definitive studies of effects on biological properties have been completed. To achieve reproducible clinical results and to permit the determination of the effects of properties of the coating on biological performance, it is essential that the properties of both clinical and experimental materials be well-characterized and consistent.
3.3 This practice provides procedures for determination of the percentage by weight of the crystalline phases identified as hydroxyapatite, β-TCP and CaO in plasma-sprayed hydroxyapatite coatings.
amorphous calcium phosphate (ACP); hydroxyapatite coatings; hydroxyapatite (HAP); hydroxylapatite coatings; phase analysis; tricalcium phosphate; whitlockite;; ICS Number Code 25.220.99 (Other treatments and coatings)
To find similar documents by ASTM Volume:
To find similar documents by classification:
This document comes with our free Notification Service, good for the life of the document.
This document is available in either Paper or PDF format.
2010 R16 EDITION
Oct. 15, 2016