ASTM-F2503 › Standard Practice for Marking Medical Devices and Other Items for Safety in the Magnetic Resonance Environment
Show Complete Document History
The following bibliographic material is provided to assist you with your purchasing decision:
Changes from the previous issue
A redline edition is available for this document, with all changes visible. Ask Document Center Inc. for more information.
1.1 This international standard applies to the practice of marking of items that might be used in the magnetic resonance (MR) environment.
1.2 The purpose of this practice is to mark items that might be brought into the MR environment and to recommend information that should be included in the marking.
1.3 The standard specifies the permanent marking of items, which are used in an MR environment, by means of terms and icons.
1.4 MR image artifacts are not considered to be a performance issue and so are not addressed in this international standard practice (see X1.5).
1.5 The values stated in SI units are to be regarded as standard. No other units of measurement are included in this standard.
1.6 This standard does not purport to address all of the safety concerns, if any, associated with its use. It is the responsibility of the user of this standard to establish appropriate safety and health practices and determine the applicability of regulatory limitations prior to use.
Significance and Use
4.1 Interactions of medical devices and other items with the MR environment has resulted in serious injuries and death of patients and other individuals. Additionally, hazards stemming from equipment malfunction are of concern. Section 4.2 lists possible direct and indirect causes of hazards in the MR environment.
4.3 This practice provides a uniform system for marking to indicate the conditions for which it has been determined that a medical device or other item may be safely placed and used in the MR environment. It provides simple visual icons and terms which are intended to reduce injuries and other mishaps that occur when items that pose hazards in the MR environment are brought into the MR environment.
magnet; magnetic; medical devices; metals (for surgical implants); MRI (magnetic resonance imaging); MR safety; ICS Number Code 11.040.01 (Medical equipment in general)
To find similar documents by ASTM Volume:
To find similar documents by classification:
This document comes with our free Notification Service, good for the life of the document.
This document is available in either Paper or PDF format.
July 15, 2013